Guzilkree A Lummer-Brodhun cube is used, as it is more sensitive than any other type of screen and permits a quicker and more accurate balance. An index point mscbeth attached to the bottom of the tube. Assuming that the illuminometer has been calibrated, place the test plate at the point in the plane horizontal or inclined where a measurement is desired. A direct and simple means is provided for standardizing the Mazda working standard lamp within the instrument.
|Published (Last):||11 November 2013|
|PDF File Size:||11.42 Mb|
|ePub File Size:||6.13 Mb|
|Price:||Free* [*Free Regsitration Required]|
Guzilkree A Lummer-Brodhun cube is used, as it is more sensitive than any other type of screen and permits a quicker and more accurate balance. An index point mscbeth attached to the bottom of the tube.
Assuming that the illuminometer has been calibrated, place the test plate at the point in the plane horizontal or inclined where a measurement is desired. A direct and simple means is provided for standardizing the Mazda working standard lamp within the instrument.
This device was used as a portable illuminometer. Connect the battery to the leads provided. All these parts are mounted upon a hard-rubber plate, which is in turn mounted upon a hardwood base. Measure the illumination upon the test plate in the manner previously described.
Lumen The lumen is the unit of luminous flux or emanation of light from a source, and is that quantity of light which will produce a normal illumination of one foot-candle over a surface llluminometer one square foot. Connect the Reference Standard to the receptacle marked A and the Illuminometer to the receptacle marked B, using the flexible cords provided for this purpose.
When using these screens illuminomete must be used in interpreting the results. If too low the screen should be placed in Slot 2. The prisms are so arranged that a circular field, and not the illuminometfr elliptical field, is seen when looking through the telescope. View the test illumknometer through the telescope, turning the knurled handle on the lower end to shift the working standard nearer to or away from the prism until a balance is secured, i. The convenience and compactness of the Macbeth Illuminometer have resulted in its adoption as a part of other photometric instruments.
The test plate becomes a secondary source of light, the brightness of which is compared with that of a translucent screen within the instrument which is illuminated to a known intensity by a small Mazda lamp. If it is desired to reduce these readings to values in units of candle-power per square foot, divide by t or 3.
A new current value will be required for the working standard. The mean of a number of measurements from a number of operators gives a precise estimate of brightness. Illumionmeter the equality of brightness type of Lummer-Brodhun screen when looking through the telescope there will be seen two concentric fields.
Colour matching filters for Macbeth illuminometer Right click equation to reveal menu options. The accuracy of the Macbeth illuminometer was studied by measuring a known source. Showing Horn Attached to llluminometer SCREENS Continued Special screens of low absorption, which may be used together with any of the neutral ab- sorbing screens, are available for use in making measurements of daylight, etc.
When using a screen in Slot 1 the scale reading should be multiplied by the factor given in the certificate for the screen. The scale of the instrument is calibrated in foot-candles, and when standardized it allows illyminometer the absorption of the test plate; hence the device is direct reading and the values secured give the intensity of illumination on a given surface.
Login or Create Account. Cited By You do not have subscription access to this journal. With the Macbeth Illuminometer as ordinarily used, that is, placing the illuminometrr plate in the plane where measurements are being made, the operator can stand upright in a normal position when making tests. Any of these publications will be sent upon request.
The one marked — should be connected to the receptacle marked 2. Cross Section of Reference Standard In the third element, or Reference Standard, is presented a decided improvement in illuminometer design.
Article tables are available to subscribers only. That this has been successfully done is attested by the fact that the scale follows the inverse square law.
Figures 7 You do not have subscription access to this journal. Holding the illuminometer illuminmoeter, sight through the two slits near the bottom of the tube and adjust the rack illuminomeger and down until the center of the filament is in line with the centers of the slits. Throw mabceth double-throw switch to Band, placing the sighting aperture of the illuminometer into the hole D Jacbeth.
By the use of the Reference Standard, the illuminometer may be macneth at any ,acbeth or place, doing away with the necessity of a dark room and auxiliary photometric apparatus and with the inconvenience and expense of laboratory standardized working standard lamps. Turn the rheostat A in the direction opposite to that indicated by the arrow, until the reading of the mil-ammeter corresponds to the current value on the Reference Standard certificate.
Tripod Attachment This is an attachment provided for holding the illuminometer in a horizontal position on the tripod when it is desired to use the instrument for making illumination measurements with the horn and translucent test plate at the inch height. Colour matching filters for Macbeth illuminometer — IOPscience Care has been used in the selection of this plate to insure, as far as possible, its coefficient of reflection being constant for all illumunometer of incident light.
Each screen has its factor certified, but these factors hold only for the particular color of illumination for which the screen is intended. There is now an illumination upon the test plate corresponding to the value given in the certificate. At 30 feet the diameter illukinometer the field is 3 feet. An important advance, however, has been made beyond previous practice. This intensity varies inversely as the square of the distance of the surface from the source.
Bureau of Standards by Mr. Related Posts
MACBETH ILLUMINOMETER PDF
When using a screen in Slot 1 the scale reading should be multiplied by the factor given in the certificate for the screen. Table I Percentages of total brightness measurements falling within deviation ranges. With the equality of brightness type of Lummer-Brodhun screen when looking through the telescope there will be seen two concentric fields. This renders the scale direct reading throughout its entire range for the new lamp. By the use of the reference standard, personal errors are eliminated, for each observer makes his own calibration of the illuminometer. If, however, he should desire to make measurements in the old way, a cap for mounting on the horn and carrying a small translucent test plate is furnished with the equipment.
It comes in a custom suitcase with compartments fitted for the various parts. The SEI Photometer dates from and replaces the inverse square law Wiki rack and pinion slider with a variable graduated neutral density filter. Description When I took photography classes at De Anza College the black and white class made use of The Zone System Manual by Minor White Wiki and for that class I got a Calumet 4x5 view camera and custom made a rectangular top plate for the Tiltall tripod to match the bottom rail mount of the camera. The spot in the center of the field of view changes brightness. There is a battery and a calibrated lamp along with an adjustable potentiometer. The pot is calibrated in light values so when the spot is the same brightness as the surrounding part of the scene you know the exposure for that spot. You could come up with film speeds for "pushed" processing as well.