Tauzilkree You need lots more information about ashm equipment such as baud rate and handshaking. Extended error recovery features round up this full-featured, comfortable protocol driver. Then you have no option but to get the information from the manufacturer concerning how to initiate a data transfer. So i got a professional to make the exact cable which is required by the device manufacturer and I am seeing a different error now. The Logging facility logs the data exchange including the control characters to a file.
|Published (Last):||24 January 2011|
|PDF File Size:||8.57 Mb|
|ePub File Size:||13.45 Mb|
|Price:||Free* [*Free Regsitration Required]|
You are on page 1of 7 Search inside document Designation: E 95 An American National Standard Standard Specication for Low-Level Protocol to Transfer Messages Between Clinical Laboratory Instruments and Computer Systems 1 This standard is issued under the xed designation E ; the number immediately following the designation indicates the year of original adoption or, in the case of revision, the year of last revision.
A number in parentheses indicates the year of last reapproval. A superscript epsilon e indicates an editorial change since the last revision or reapproval. Scope 1. The clinical laboratory instruments under consideration are those that measure one or more parameters from one or more patient samples.
Often they will be automated instruments that measure many parameters from many patient samples. The computer systems considered here are those that are congured to accept instrument results for further processing, storage, reporting, or manipulation. This instrument output may include patient results, quality control results, and other related information.
The electrical and mechanical connection between instrument and computer is described in the Physical Layer section. The methods for establishing communication, error detection, error recovery, and sending and receiving of messages are described in the Data Link Layer section.
The data link layer interacts with higher layers in terms of sends and receives messages, handles data link connection and release requests, and reports the data link status. The major topics are found in the following sections: Section.
Scope 1. The clinical laboratory instruments under consideration are those that measure one or more parameters from one or more patient samples. Often they will be automated instruments that measure many parameters from many patient samples. The computer systems considered here are those that are configured to accept instrument results for further processing, storage, reporting, or manipulation.
ASTM E 1381-95 PDF
Faegar All communicationparameters like baud rate, number of data bits, etc. You currently cannot view this content. It is unlikely that most of it will work in the same manner, and even when get it communicating you still have to decode the data it is transmitting. Monday, December 5, 9: Well it could be possible the cable you had made is not correctly made. Object, e As System.